Biomolecules and nanostructures
The Optical Sciences group studies the interaction of light and matter. Our current focus is on detection and sensing/imaging with an emphasis on the development of integrated photonics. We are part of Twente University's Department of Science and Technology and member of the MESA+ institute.
Combined near field optical and atomic force microscope
This instrumentation oriented FOM-BR project started in 1993 and will be concluded with a PhD thesis on 19th June 1997.
An integrated NSOM/AFM probe, which exists of an 200µm long flexible cantilever with at its end a 5*m high conical tip, has been realised in the clean room facility of the MESA research institute using standard micro-mechanical techniques like wet and dry etching, photolithography and CVD methods on a 3" Si-wafer. Furthermore a combined NSOM/AFM set-up (both transmission and reflection) has been built based on a static light path and probe and a scanning sample stage. The shape of the probe (cantilever flexibility and tip shape) is suitable for use in AFM applications. Most effort is put in the deposition of an opaque layer surrounding the tip in order to get a sub-micron aperture at the tip end. Aluminium has been chosen as coating material because of its high extinction coefficient and well known depositing characteristics. Because of tip shape and oxidation properties of the Aluminium other coating materials have been tried also. The optical resolution found so far, ~ 300 nm, is not yet competitive with optimised aperture type fibre probes.
Shear force detection using a quartz crystal tuning fork has been implemented to provide higher topographic sensitivity, lower force interaction, reduced optical background signal and liquid operation. This shear force feedback system is used as distance control mechanism for a near-field scanning optical microscope, in which a tapered optical fibre attached to the tuning fork is scanned over the sample surface. The dynamics of a tuning fork shear-force feedback system have been investigated in detail. Experiments were performed measuring amplitude and phase of the tuning fork oscillation as a function of driving frequency and tip-sample distance. These experiments reveal that the resonance frequency of the tuning fork changes upon approaching the sample. Either amplitude or phase of the tuning fork can be used as distance control parameter in the feedback system. Using amplitude a second-order behaviour is observed while with phase only a first-order behaviour is observed. Numerical calculations confirm these observations. This first-order behaviour results in an improved stability of our feedback system. As an example a sample consisting of DNA strands on mica was imaged which showed a height of the DNA of 1.4 nm.
The following articles have been published regarding this project:
Atomic steps with tuning-fork-based noncontact atomic force microscopy(abstract) (full pdf)
Rensen WHJ, van Hulst NF, Ruiter AGT, West PE
APPLIED PHYSICS LETTERS
vol 75 issue 11: p1640-p1642 SEP 13 1999
Tuning fork shear-force feedback(abstract) (full pdf)
Ruiter AGT, van der Werf KO, Veerman JA, Garcia-Parajo MF, Rensen WHJ, van Hulst NF
vol 71 issue 1-4: p149-p157 MAR 1998
Single molecule rotational and translational diffusion observed by near-field scanning optical microscopy(abstract)
Ruiter AGT, Veerman JA, GarciaParajo MF, vanHulst NF
JOURNAL OF PHYSICAL CHEMISTRY A
vol 101 issue 40: p7318-p7323 OCT 2 1997
Dynamic behavior of tuning fork shear-force feedback(abstract) (full pdf)
Ruiter AGT, Veerman JA, vanderWerf KO, vanHulst NF
APPLIED PHYSICS LETTERS
vol 71 issue 1: p28-p30 JUL 7 1997
Microfabrication of near-field optical probes(abstract)
Ruiter AGT, Moers MHP, vanHulst NF, deBoer M
JOURNAL OF VACUUM SCIENCE & TECHNOLOGY B
vol 14 issue 2: p597-p601 MAR-APR 1996
Development of an integrated NSOM probe(abstract) (full pdf)
Ruiter AGT, Moers MHP, Jalocha A, vanHulst NF
vol 61 issue 1-4: p139-p143 DEC 1995